The 1J1016 SMD 2-Terminal High Current Fixture is used to connect a Wayne Kerr Analyzer (3255B or 3260B) and DC Bias Unit (3265B) system to a surface mount Device Under Test and pass up to 125 A DC bias current.
The 1J1016 Fixture can be used with the following systems:
DC Bias Unit
Maximum measurement frequency
Maximum DC bias current
125 A using 5 units in parallel
50 A using 2 units in parallel
20 Hz to 3 MHz
DUT Max Temperature:
200 ºC for 1 hour
The measurement leads are connected to the analyzer (3255B/3260B) front panel BNC’s.
The high current leads are connected to the high current terminals of the 3265B DC Bias Unit.
2-terminal connection to the bottom face of Device Under Test.
When the fixture cover is opened, the safety interlock will operate and stop the DC bias current.
185 mm × 90 mm × 190 mm (L x W x H)
DC Bias Current Unit, To evaluate components at high currents the 3265B/25A DC Bias Units may be used with either the Wayne Kerr 3255B Inductance Analyzer or the 3260B Precision Magnetics Analyzer. Bias current may be set from 25mA to 25A in steps of 25mA.
Precision Magnetics Analyzer, Offers all measurement parameters needed for the analysis of magnetic components up to 3MHz including true 4-terminal primary to secondary parameters by using 8 front panel BNC’s.
Precision Magnetics Analyzer, Highly versatile and accurate inductance analyzers able to characterise devices in a clear and simple manner